Breath Alcohol Analyzer is a microprocessor based handheld instruments, which measures the blood alcohol content in a person who has consumed alcohol. It is done by measuring the breath alcohol content and relating it to the blood alcohol content.
When the breath sample is delivered, the software checks the moderate blow of breath sample continuously for 5 seconds and then the instruments draws approximately 200 µl of breath into the fuel cell by means of a pump in 200 miliseconds.
The instrument is calibrated on mg/100ml or BAC scale. The BAC unit defines blood alcohol content in terms of grams of alcohol in 100ml of Blood, for example: 0.05BAC means 0.05 grams of alcohol in 100ml of blood. With DOT MATRIX Printer
Complete with: Carrying case, 20 Nos Mouth pieces, Data logging Cable, Charger User Manual and Calibration certificate.
|Gas Related Specifications
|0 – 500 mg / 100 ml of blood
|1 mg / 100 ml of blood
|Electro Chemical Sensor
|+/- 5 mg/100 ml at 100 mg/100ml
|By actively blowing through specially designed mouthpiece
In-built pressure sensor should give visual indication of
inadequate breath sample
|Blowing over the sensor or exposing the sensor to breath of involuntary subject without using the mouthpiece
|Display & Indicator Specifications
|Backlit LCD Display
|LEDs to indicate status of Power
|Clear indication should be given by means of Red LED and
audible buzzer for Positive Result for easy understanding
of the user
|For Setting, Calibration, Test and Download
|Microprocessor based instrument with Real Time Clock and
|Within 3 seconds
|If negative results,recovery time -5 seconds, If positive results, recovery time -30 seconds.
|Data logging facility for minimum 3000 samples in its
non volatile in built memory
|Should record test result with date and time
|Dot Matrix Printer with built in rechargeable battery capable
of at least 300 printout with a single charge
|Rechargeable Li Ion battery
|Should print Instrument serial no., last calibration date,
est date and time, Alcohol Content and Result – negative or positive.
Any other details as required by the Dept. should be
incorporated in the printout.
|Report Format Customization
|Test Report should be customizable as per
Dept.’s requirements , LIKE the top line of the print
out should indicate “DRUNKEN DRIVING CHALLAN “ ETC.
|Operating Temperature Range
|0 to +50 deg C
|Ambient +/- 10%
|500 g or less
|Hand held, Compact
|Instrument should be calibrated using traceable gas standards. Calibration unit is to be shown to this office along with demo
and calibration is required every 4 months by the supplier
|Standard Accessories supplied with the Instrument
Battery Charger for Printer and Analyzer,
Mouthpieces 20 numbers
printer paper rolls
PC Data Transfer Cable
© 2016 Websoft Solution. All Rights Reserved | Designed by Infosky Solutions